To perform a proper duct leakage test using an ORIFLOW duct leakage tester, perform the following
- Use the orifice plate with the recommended leakage range for your job. Use our online program to help determine which plate(s) you can use with our testers.
- Make sure you have adequate electrical power for the fan. DO NOT OVERLOAD extension cords. Make sure you use one with the proper gauge.
- Cover all outlets with plastic bags and seal with duct tape.
- Allow all joints and seams that were sealed with duct sealer to cure for at least 24 hours or the required cure time by the manufacturer, whichever is greater.
- If possible, position the ORIFLOW duct leakage tester in a location where there will be a straight path from the orifice tube outlet to the location where you will feed air into the system.
- Drill a 3/8″ diameter hole no closer than two or three feet away from the flexible duct to systemjoint.
- Place the pressure tubing into the previously drilled hole; extend 6 to 12 inches into the duct.
- Seal the tubing penetration joint with putty or tape.
- Zero both gauges. Locate adjusting screw on center front of gauge. Use a small slotted
screwdriver. Turning clockwise increases the pressure reading, counter-clockwise decreases it.
- Close the fan’s inlet damper so the duct system does not get overpressurized by the tester.
- Turn on the tester blower.
- Adjust inlet damper until the reading on the gauge marked “Duct System” is at the system pressure required for performing the leakage test.
- Once the duct system pressure has been reached and stabilized, note the reading on the
gauge marked “Orifice Tube”. This pressure drop reading will correspond to the amount of air
you’re feeding into the duct system, which is the amount of air that is leaking out.
- Refer to the calibration table that came with your test rig. Find the pressure reading and note the corresponding leakage rate.
If the “Orifice Tube” gauge exceeds the maximum range, the system is probably leaking too much air. Investigate system for missed outlets or unsealed joints. You can also use a larger orifice or a gauge with a larger range
If you cannot obtain any pressure (or very little pressure) with the blower on and the inlet damper wide open, it could be due to one of the following problems:
- Using too small of an orifice plate
If pressure drop across orifice plate is more than 2 in.wg., use the next greater sized
orifice plate. Use our online program to determine which orifice plate you should use
based on the maximum allowable leakage. You can also refer to graphs for each
- System has more leakage at test pressure than is allowed
When using the Cobra or Lynx tester, use the 4-inch plate to pressurize your system
and refer to the following table to determine if your system is leaking more than the
maximum capacity of the tester.
|Maximum Leakage Capacity (cfm)*|
|Model||0.1 in.wg.||1 in.wg.||2 in.wg.||4 in.wg.||6 in.wg.||7 in.wg.||8 in.wg.||9 in.wg.||10 in.wg.||12 in.wg.||14 in.wg.||16 in.wg.|
|Lynx||570 cfm||550 cfm||540 cfm||455 cfm||290 cfm||160 cfm||—||—||—||—||—||—|
|Cobra||680 cfm||650 cfm||645 cfm||600 cfm||465 cfm||405 cfm||350 cfm||250 cfm||—||—||—||—|
|Panther||1400 cfm||1350 cfm||1320 cfm||1210 cfm||1065 cfm||970 cfm||875 cfm||745 cfm||615 cfm||270 cfm||—||—|
|Rhino||1700 cfm||1630 cfm||1550 cfm||1420 cfm||1330 cfm||1255 cfm||1185 cfm||1110 cfm||1035 cfm||895 cfm||660 cfm||480 cfm|
* – Using the largest size orifice plate. Note each tester comes with one plate. Other orifice plate sizes are available for accurate testing of different flow ranges. Use smaller orifices for smaller flow ranges; largest orifice for maximum flow range.